I tried to use this driver on laptop with integrated IRda and winXP to connect glucometer ACCU-CHEK,
but after installation I get virtual port with number 7.
software I use, serve only COM-1 to COM-3.
Does anybody know how to change this settings for "lower" com port ?
You should disable the COM-1 in the peripheric manager Windows and then re-install IrCOMM2k.
read the manual....
In order to manually specify the port, start regedt32.exe (e.g. via "Run" in the Start Menu). Below Registry-entries have to be adjusted, some require additional access rights:
(should the desired COM port number be missing altogether, please add it)
(Cosmetic, so that the device manager displays a meaningful name)
When above changes were made, the virtual COM port has to be disabled, then re-enabled in the Device Manager (or, alternatively, restart the entire system).
Log in to post a comment.