While using owfs-2.8p20 with a Hobby Boards Moisture Meter I noticed 3 problems.
1) OWFS does not present a "calibration" directory under moisture.
2) "moisture/calibrate/max" actually modifies the minimum value.
2) OWFS has the "LEAF Wetness" and "Soil MOISTURE" modes reversed. Per HB user
manual dated 5/21/12 Leaf Wetness should range from 0-100. Soil Moisture ranges
from 0-199. OWFS accesses them reversed. I have included a diff file with my
proposed fixes for ow_eeef.c.
Best Regards, RichM