On Wed, May 25, 2011 at 11:40 AM, Nico Stuurman <nico@...:
> We have no such scope to test. However, I added a simple check in
> ZeissCAN29.cpp in the Turret object. It should simply not send a move
> command and not set the microscope to be busy. Can you check it out of the
> repository and let me know what you find?
> Also, I am a bit worried that other devices may behave the same behavior.
> Please let me know of such problems.
The fix appears to have worked. No Timeout issue with uManager 1.4.
I couldn't get the andor lib to load since the Linux SDK seems to be behind
the windows SDK once again, so I will be moving back to uManager 1.3 until
Andor updates the Linux SDK update.
I also have a related or unrelated timeout with the XY Stage. I am still
trying to determine if it is a hardware issue or a software issue. I see a
XYStage time out around ~column 6 of a well plate. Physically bumping the
manual XY Stage controller when I see this happens prevents a uManager
timeout. This is a bit harder to reproduce and nail down.
Biomedical Imaging Group
University of Massachusetts Medical School